Surface mount components
Dan Michaels email (remove spam text)
>> Yeah, who/what can be trusted at all anymore? I put my sine
>> generator to the cap marked 220 thru a 10K [5%] resistor, and
>> adjusted freq until the voltage across the cap dropped to 50%
>> on the scope. Using Xc = 1/(2*pi*F*C)
>That's the magnitude of the impedance, but is not valid the way you used it
>due to being 90 degrees out of phase with the resistor.
Thank you - as was also brought to my attention by RichO.
[I am now limping off into the sunset on my single living brain
http://www.piclist.com hint: To leave the PICList
See also: www.piclist.com/techref/index.htm?key=surface+mount+components
You must be a member of the
piclist mailing list
(not only a www.piclist.com member) to post to the