Searching \ for '[PIC:] want to make a LC meter with pic.' in subject line. ()
Help us get a faster server
FAQ page: www.piclist.com/techref/microchip/devices.htm?key=pic
Search entire site for: 'want to make a LC meter with pic.'.

Exact match. Not showing close matches.
'[PIC:] want to make a LC meter with pic.'
2004\01\15@120041 by

f sweeping, serialing and comparing with a R, is a excurate method.
any exprience to share, by using PIC?

Or, any other method to do it? lots of multimeter has C or CL meter.
what method do they use?

--
http://www.piclist.com#nomail Going offline? Don't AutoReply us!
email listservmitvma.mit.edu with SET PICList DIGEST in the body

Some of the inexpensive C meters I've seen have a wien bridge oscillator
generating a sine wave. This drives one of the capacitor terminals. The
other capacitor terminal drives an op-amp current to voltage converter (op
amp with resistor from inverting input to output, non-inverting input
grounded, input applied to inverting input directly). Capacitor current
with known voltage and known frequency is proportional to capacity. So,
they just use their AC voltage measurement capability to measure the
voltage at the output of the current to voltage converter.

They can't tell if it's a capacitor, a resistor, or an inductor, but it's
a simple way of measuring a capacitor if it is indeed a capacitor.

If you want to get fancy, you could generate a sine wave with a lookup
table and D/A to drive the capacitor. Then, instead of just measuring the
RMS current, do fast D/A samples of the current (as indicated by the
current to voltage converter). Do a complex number division (using either
polar or rectangular notation, whichever works out best), to get Z=V/I . Z
can then be represented as R+jX . Knowing the frequency, you can convert
from X to inducatance or capacitance and model the unknown impedance as
L/C in series with R or L/C in parallel with a different R.

Another approach for use at RF that I think would be fun to try would be
to use a reflectometer and measure the voltage ratio and phase between the
forward and reverse samples. I believe these could then place the load
impedance on a Smith chart where the impedance could be read directly.

Harold

> f sweeping, serialing and comparing with a R, is a excurate method.
>
> any exprience to share, by using PIC?
>
> Or, any other method to do it? lots of multimeter has C or CL meter.
> what method do they use?
>

--
FCC Rules Online at http://www.hallikainen.com

--
http://www.piclist.com#nomail Going offline? Don't AutoReply us!
email listservmitvma.mit.edu with SET PICList DIGEST in the body

http://www.freddospage.nl/

http://www.antrak.org.tr/gazete/111998/barbar.htm

http://www.elektroda.pl/eboard/ftopic37668-0-asc-30.html

http://www.spettel.de/ralf/projekte/lc-meter/index.html

And these ones using an ATMEL processor:

http://xavier.fenard.free.fr/LCMeter.htm

http://www.hw.cz/constrc/lc_metr/

Mark PY3SS

On 15 Jan 2004 at 12:02, paul x wrote:

{Quote hidden}

--
http://www.piclist.com#nomail Going offline? Don't AutoReply us!
email listservmitvma.mit.edu with SET PICList DIGEST in the body

Check out practical electronics there is one in the feb issue

Steve...

-----Original Message-----
From: pic microcontroller discussion list
[PICLISTMITVMA.MIT.EDU] On Behalf Of paul x
Sent: 15 January 2004 17:02
To: PICLISTMITVMA.MIT.EDU
Subject: [PIC:] want to make a LC meter with pic.

f sweeping, serialing and comparing with a R, is a excurate method.

any exprience to share, by using PIC?

Or, any other method to do it? lots of multimeter has C or CL meter.
what method do they use?

--
http://www.piclist.com#nomail Going offline? Don't AutoReply us!
email listservmitvma.mit.edu with SET PICList DIGEST in the body

--
http://www.piclist.com#nomail Going offline? Don't AutoReply us!
email listservmitvma.mit.edu with SET PICList DIGEST in the body

{Quote hidden}

An analog method would be to generate sine and cosine waves with
with samples of both input phases will give you DC voltages which tell you
the real and imaginary parts of the impedance (actually the admittance).

John Power

--
http://www.piclist.com hint: To leave the PICList
piclist-unsubscribe-requestmitvma.mit.edu

The last EPE magazine contains a 16F628-based LCF meter. Uses an LC
oscillator and measures the frequency.

Wouter van Ooijen

-- -------------------------------------------
Van Ooijen Technische Informatica: http://www.voti.nl
consultancy, development, PICmicro products

--
http://www.piclist.com hint: To leave the PICList
piclist-unsubscribe-requestmitvma.mit.edu

>
> An analog method would be to generate sine and cosine waves with
> output
> with samples of both input phases will give you DC voltages which tell you
> the real and imaginary parts of the impedance (actually the admittance).
>
> John Power
>

Nice solution! It should be pointed out that "mixing" is analog
multiplication followed by a low pass filter to yield the DC component of
the product.

A quadrature output oscillator can be constructed with (as I recall) three
op amps and a bunch of resistors and capacitors. I dont' think it's very
easy to change the frequency, though. Direct Digital Sysnthesis is a nice
way to generate the quadrature signals, though not especially cheap
(unless you use a couple D/As on a PIC with a lookup table, which would be
somewhat frequency limted).

Harold

--
FCC Rules Online at http://www.hallikainen.com

--
http://www.piclist.com hint: To leave the PICList
piclist-unsubscribe-requestmitvma.mit.edu

Here's one, pic based.  The construction was detailed in an older Radio and
Electronics issue.  I 'think' he has the schematic somewhere on his website.

Dave

--
http://www.piclist.com hint: To leave the PICList
piclist-unsubscribe-requestmitvma.mit.edu

Hi!

There is one on the latest (February) issue of EPE magazine.

http://www.epemag.com

Cheers,
Diego.

--